Firstly, Point of Care Analyser for Hormone Tests, Vitamins Testa, Tumour Tests, Thyroid Tests, Renal Tests, Cardiac Tests, and Diabetes Tests in less than 30 mins. Fluoroimmunoassay Analyser.
Point of Care Analyser – Results in 20 mins
Therefore, the Dry Fluoroimmunoassay Analyzer represents advanced technologies of the same type. Also, it can conveniently, accurately, and quickly test relevant parameters of the human body when used cooperatively with fluoroimmunoassay reagent strips.
Dry Fluoroimmunoassay Analyzer uses a LED as the excitation light source. The light from the LED hits a test device that has been inserted into the meter. This causes the fluorescence dye in the test device to give off energy. The more energy the fluorescent dye gives off, the stronger the signal.
And so, after a buffer-mixed sample is added to the test device strip, the test device is inserted into the Dry Fluoroimmunoassay Analyzer.
Therefore, see our Comparison Data Point of Care Instrument.
Also, Medicines Online Laboratory performs comparison tests to compare the results on the same samples for multiple UK laboratories. Therefore, the results are these comparison tests are available upon request. But, compliance testing and comparison testing are part of our Laboratory function. So, we are not merely a wholesaler. Additionally, being a Diagnostic Laboratory, we have access to a quality control team to conduct performance measurements and controls.
Whole blood, serum, plasma, urine, and faeces (subject to the specification of the immunofluorescence assay kit)
The variable coefficient of fluorescence value of fluorescence test cards of different intensities repeatedly measured by the instrument CV≤2.0%
The relative deviation between the 4th hour and the 8th hour after the instrument is working stable, and the test result when the device just works stably is limited to ±3.0%.
When the instrument tests the calibrated fluorescence test card, the deviation between the measured fluorescence value and the calibration value of the fluorescence test card is ±3.0%.
The time from inserting the fluorescence test card to displaying the test result should not exceed 15s.
The linearly dependent coefficient of the fluorescence value and concentration read by the instrument r 0.990